I machined a delrin insulator with fiberglass center rod for the CF 4 pin electrical feedthrough to provide power to the two incandescent substrate heaters in my new sputtering setup and to allow for instrumentation hookup.
New sputtering setup has dual substrate heaters as well as insulated feedthroughs for both target and substrate so that it can operate in DC, RF, Bias sputtering modes, or any combination therein. A quartz microbalance thickness monitor will be added soon.
The second to last image shows multiple colors in glow discharge during sputtering possibly due to DC bias. Bottom and top electrodes are both at different potentials with respect to chamber ground, so it is possible that different gasses are ionized due to differences in ionization energies of the residual gaseous species.
A Faraday cup is an instrument that can be used to measure ion beam currents. I designed this one for testing the anode layer ion source from rtftechnologies.org. It has 2 permanent magnet rods (3/16″ dia x .5″) in the first part of the cup to act as a trap for incoming electrons and the cup can be positively biased to suppress secondary electrons so that the measured current is just the ion beam.
Between the section with the magnets and the collector, there is an interchangeable aperture so you can use different sizes. This was designed to be hand made on a lathe; there is no tolerancing so do not attempt to make it unmodified on a CNC. The body, aperture, and collector are made from stainless steel and the ring which holds the magnets and insulates the collector from the body is plastic (ideally PEEK). The backside of the cup has two blind holes, the center for mounting and the offset hole for BNC connection.
My vacuum chamber which I am setting up for thermal evaporation has a large rectangular flange which has grooves for ISO style clamps and 2 locator pins. This flange may have been used for something like a load lock but as far as I could tell it is not a standard size so a custom block off plate was designed and machined from SS with grooves on the back for clamps, 2 pressed locator pins, and a viton o-ring. Mating surface was hand lapped to ensure good vacuum seal. Made by the extremely talented J.D Guerard machining company.
Drilling and tapping 1/8 NPT in a 2.75″ CF blank flange to mount a Hastings Instruments DV-23 thermocouple gauge. This, along with a Baratron pressure (capacitive) transducer, will be my main low to mid vacuum gauge.